Accurate1 & fast results the easy way
Auto-coding: Inserting just a chip makes testing simple right from the start. Simple auto-coding, Strip expiry warning, Wide operating temperature & Humidity range
Very easy and user-friendly handling: A large display, simple handling with just two buttons and the possibility of out-of-meter dosing
Accurate1 and fast results: The meter requires no more than 5 seconds to measure and delivers results comparable to a reference method.
State-of-the-art technology for reliability
Under-dose detection: Signals application of too little blood to test strip
Strip expiry warning: Alerts users and marks results if test strips are past their expiry date
Visual back up: The test strip itself allows user to quickly double-check test results by using the colour coded visual found on the strip vial
State-of-the-art diabetes management tools
The Accu-Chek Active system now offers functions to mark and evaluate the pre- and post-meal blood glucose values.The new Accu-Chek Active system provides the option to flag thses values of particular importance and thus supports a precise evaluation of the metabolic situation.
Furthermore, the averages of pre- and post-meal values for the last 7, 14 or 30 days can be retrieved from the meter's memory and displayed directly on its screen.
Easy-to-use data analysis function: Accu-Chek Active has a 350-value memory for saving results together with date and times as well as an infrared interface for communicating with a PC running a suitable analysis program.
Fast five second test time: With a testing time of about 5 seconds, your Accu-Chek Active is one of the fastest blood glucose monitors available.
Ergonomic design: Your new Accu-Chek Active is designed for ergonomic functionality with good looks.
Small blood sample: Your Accu-Chek Active requires only a small drop of blood, 1 to 2μL if you apply the blood to the middle of the test pad. If at any time the drop you apply is too small, the monitor will detect this and issue a warning.
Easy to operate: No need to press buttons when you carry out the test.
Out-of-meter dosing: After activating the test cycle you may remove the test strip to apply blood (eg. from an earlobe) and then insert the strip again in the monitor.
Accurate1 results: Application of blood to the test strip starts a colour reaction. The final colour is accurately read(measured photo metrically) by the meter's optical system and the result is converted to a blood glucose value.
Auto Coding: Each pack of test strips contains a coding chip that holds all of the important informtion that is needed to evaluate the strips from that particular pack. Whenever you open a new pack of strips, simply insert the new coding chip in the coding chip slot.
Easy to read display: Numbers and messages are clearly presented on a spacious display screen. Easily understood symbols give you additional information and instructions.
Virtually pain-free testing in 5 seconds.
Model: Accu-Chek Active (Model GC)
Measuring range: 0.6 mmol/L - 33.3 mmol/L
Blood volume: 1 - 2 μL
Battery: 1 battery (type CR 2032)
Batterylife: Approximately 1000 tests or approx. one year
Auto power off: after 30 or 90 seconds depending on operating status
Memory capacity: Up to 350 measurements with date and time
Test averages: 7, 14, 30 days
Data transfer: Via infrared interface (wireless) to supported Accu-Chek software
Display: 7 segment LCD with symbols
Under-dosing detection: Yes
Automatic coding: Via reliable code chip
Approx. 5 sec (blood application with test strip within the meter)
Approx. 10 sec (blood application with test strip outside the meter)
Measuring principle: Photometric determination of glucose by means of glucose-dye-oxidoreductase. (synonym: PQQ-dependent glucose dehydrogenase mediator reaction)
Measuring conditions: Temperature: +10°C to +40°C (+50°F to +104°F)
During testing - up to 85 % relative humidity
During storage - up to 93% relative humidity
Without battery: -25°C to +70°C (-13°F to +158°F)
With battery: -10°C to +50°C (+14°F to +122°F)
Dimensions: 104 x 52 x 21 mm
Weight: 53 g without battery
Altitude independence: 0 - 4000m (0 - 13123 feet)
HOW TO USE
How to use the Accu-Chek Active (GC)
Coding the meter
|When coding your Accu-Chek Active meter, leave it turned off. Take the orange-coloured code chip from the pack of test strips and slide it gently into the slot on the side of the meter. Be sure to push the code chip in all the way until it clicks into place. Insert a test strip to turn on the meter.|
Compare the three-digit code number (eg. 689) on the displaly with the number printed on the test strip container. The two numbers must be identical.
Testing your Blood Glucose
on the meter: Inserting the test strip automatically turns the meter
on and puts it in test mode. Hold the test strip so that the
application area and arrows are facing upwards. Gently push the test
strip in the direction of the arrows into the test strip slot of your
Accu-Chek Active until you hear it click into place.
|Out-of-meter dosing: Pull the test strip out of the meter as soon as the drop symbol flashes in the display. You now have 20 seconds to apply blood to the test strip and re-insert it before the meter switches off.|
Reading the blood glucose result and plausibility check
the blood glucose result: In the case of 'in-meter dosing' the test is
complete after about 5 seconds. The result appears in the display and
is automatically saved with the date and time.
|Plausibility check: You have the facility for verifying the result by comparing the colour of the round control window on the back of the test strip with the colour scale on the label of the test strip container. It is important that this check is performed within 30-60 seconds after blood application.|
Memory & cleaning the meter
|Memory: Accu-Chek Active saves automatically 350 results in memory together with the date and time. To turn on the meter and retrieve results from memory please press the M button briefly. In the display you will see 'memory' at top left and the most recent blood glucose result. Press the S button to move back one result in time.|
|Cleaning the meter: Slide off the test strip guide towards you and clean it with cold water. Afterwards you may disinfect the test strip guide with 70% alcohol. Wipe the measurement optics component with a soft lint-free cloth and/or cotton swab, which may be slightly moistened with water or 70% alcohol. Make sure all of the components are thoroughly dry before sliding the test strip guide back onto the meter.|
Flagging test results
You can flag test results to describe certain events connected to the result or particular characteristics of the result. You can only flag a test result while the test strip is still in the meter and the result is being displayed.
|Before meals flag
(Full apple symbol).
Intended for test results that you have measured beofre meals.
|After meal flag
(Apple core symbol)
Inteded for test results that you have measured
|General Flag (Asterisk symbol)
You can define the meaning of this flag yourself.
For example, you can use it for test results that
you have obtained from alternative sites or after physical activities.
|Control Flag (Bottle symbol)
Intended for performance checks in which
you apply control solution to the test strip
instead of blood.
For more information on how to use this meter, please refer to the Accu-Chek Active Owner's Booklet.
- Carry Case (soft pouch)
- Battery(CR 2032)
- User's manual
- Accu-Chek Softclix lancing device
- Accu-Chek lancets for virtually pain-free blood collection